0
Offline
Newbie
Karma: 1
Posts: 35
Arduino rocks


« on: November 19, 2012, 07:14:52 am » 
hello,
i'm trying to read ac current using a bidirection allegro hall sensor acs758050b i've a problem using it. when i do not connect anything to this sensor i read * a raw value of 516 with a input voltage of 4.80V +0.05V * a raw value of 512 with a input voltage of 4.70V +0.05V
to convert the value to current i use the current formula adcvref = 5 adcdiv = 1023 sensitivity = 0.04 ( acs758050b sensitivity datasheet) amp = (adcraw*adcvref/adcdiv adcvref/2) / sensitivity;
anyway the problem with the raw value persist, even if i change the reference (adcvref) current value to real adcvalue (4.80 or 4.70)
can anyone help me? thanks!



Logged





United Kingdom
Offline
Tesla Member
Karma: 224
Posts: 6614
Hofstadter's Law: It always takes longer than you expect, even when you take into account Hofstadter's Law.


« Reply #1 on: November 19, 2012, 08:13:06 am » 
The value adcdiv in your calculation should be 1024, not 1023. A value of 512 with no current flowing is exactly right, however the device may have an offset of a few mV, so the value may be a few counts out. Noise on the +5V line will affect the reading.



Logged

Formal verification of safetycritical software, software development, and electronic design and prototyping. See http://www.eschertech.com. Please do not ask for unpaid help via PM, use the forum.




0
Offline
Newbie
Karma: 1
Posts: 35
Arduino rocks


« Reply #2 on: November 19, 2012, 08:55:09 am » 
thank you for reply The value adcdiv in your calculation should be 1024, not 1023
reason for 1024, correct me if i'm wrong adcraw is a 10bit values, so from 0(min) to 1023(max = 2^101), so the max values here is 1023 A value of 512 with no current flowing is exactly right, however the device may have an offset of a few mV, so the value may be a few counts out. Noise on the +5V line will affect the reading.
which is the math to use for transform adc values to amepere using an acs758. i get adc samples for adc current making an avarage of "many" values, i.e. at least 2 times the period of my 50Hz primary current, and my adc values are stable. but maybe my formula is not correct, i'm using this one: amp = (adcraw*adcvref/adcdiv adcvref/2) / sensitivity; because with "small" adcraw changes i have "big" change in amp es. ( 512 *5/1023  5/2)/0.04 = 0.062 ( 516 *5/1023  5/2)/0.04 = 0.54 (1/2Amp!) and also consider the max raw value of 1023 ( 1023 *5/1023  5/2)/0.04 = 62.5 acs758050b is rated as 50Amp max, the max raw value i should read is almost 900



Logged





United Kingdom
Offline
Tesla Member
Karma: 224
Posts: 6614
Hofstadter's Law: It always takes longer than you expect, even when you take into account Hofstadter's Law.


« Reply #3 on: November 19, 2012, 09:31:36 am » 
reason for 1024, correct me if i'm wrong adcraw is a 10bit values, so from 0(min) to 1023(max = 2^101), so the max values here is 1023
The max value you can read is indeed 1023, but the correct value to use in the calulation is 1024. See section 24.7 of the atmega328p datasheet. i get adc samples for adc current making an avarage of "many" values, i.e. at least 2 times the period of my 50Hz primary current, and my adc values are stable.
Are you trying to measure alternating current? If so, you need to average the square of the current, and take the square root of the average (that's what RMS means). Easiest way is to subtract the zerocurrent value (512 or thereabouts) from the raw readings, square that, and add it to an accumulator variable. When you have accumulated enough samples, take the square root if the accumulator and divide by the number of readings. thank you for reply but maybe my formula is not correct, i'm using this one: amp = (adcraw*adcvref/adcdiv adcvref/2) / sensitivity;
The device is ratiometric, so its sensitivity varies with Vcc. Use: current = (adc_raw  adc_zero) * 5/(sensitivity@5V * 1024) (where adc_zero is the ADC reading you get when the current is zero, which will be close to 512), which is almost the same. because with "small" adcraw changes i have "big" change in amp es. ( 512 *5/1023  5/2)/0.04 = 0.062 ( 516 *5/1023  5/2)/0.04 = 0.54 (1/2Amp!) and also consider the max raw value of 1023 ( 1023 *5/1023  5/2)/0.04 = 62.5 acs758050b is rated as 50Amp max, the max raw value i should read is almost 900
That's the nature of the device. If you put more than 50A through it then the accuracy will degrade.



Logged

Formal verification of safetycritical software, software development, and electronic design and prototyping. See http://www.eschertech.com. Please do not ask for unpaid help via PM, use the forum.




0
Offline
Newbie
Karma: 1
Posts: 35
Arduino rocks


« Reply #4 on: November 19, 2012, 09:49:03 am » 
The max value you can read is indeed 1023, but the correct value to use in the calulation is 1024. See section 24.7 of the atmega328p datasheet.
Thank you for this info! Are you trying to measure alternating current? If so, you need to average the square of the current, and take the square root of the average (that's what RMS means). Easiest way is to subtract the zerocurrent value (512 or thereabouts) from the raw readings, square that, and add it to an accumulator variable. When you have accumulated enough samples, take the square root if the accumulator and divide by the number of readings.
Yes ac current. so, for AC current something like this samplesnum = 100; adc_zero = 512; //as read when there is no current on the load sensitivity = 0.04; //as read from datasheet for(i=0; i<samplesnum ; i++) adc_raw = analogRead(1); currentacc += (adc_raw  adc_zero) * (adc_raw  adc_zero); //subtract the zerocurrent value (512 or thereabouts) from the raw readings, square that, and add it to an accumulator variable } currentac = sqrt(currentacc)/samplesnum; //take the square root if the accumulator and divide by the number of readings. for DC current something like this adc_zero = 512; //as read when there is no current on the load sensitivity = 0.04; //as read from datasheet currentdc = (adc_raw  adc_zero) * 5/(sensitivity * 1024) right?



Logged





0
Offline
Shannon Member
Karma: 206
Posts: 12179
Arduino rocks


« Reply #5 on: November 19, 2012, 10:06:16 am » 
adc_zero = 512; //as read when there is no current on the load sensitivity = 0.04; //as read from datasheet currentdc = (adc_raw  adc_zero) * 5/(sensitivity * 1024)
That's a confusing order, I'd apply conversions onebyone: adc_zero = 512; //as read when there is no current on the load sensitivity = 0.04; //as read from datasheet amps_per_volt = 1.0 / sensitivity ; fullscale_volts = 5.0 ; fullscale_count = 1024.0 ; ampsdc = ((adc_raw  adc_zero) / fullscale_count) * fullscale_volts * amps_per_volt ;
I'd always choose a name with explicit unit if possible "current" might be amps, milliamps or microamps, whereas "amps" is unambiguous.



Logged

[ I won't respond to messages, use the forum please ]




United Kingdom
Offline
Tesla Member
Karma: 224
Posts: 6614
Hofstadter's Law: It always takes longer than you expect, even when you take into account Hofstadter's Law.


« Reply #6 on: November 19, 2012, 10:27:07 am » 
so, for AC current something like this samplesnum = 100; adc_zero = 512; //as read when there is no current on the load sensitivity = 0.04; //as read from datasheet for(i=0; i<samplesnum ; i++) adc_raw = analogRead(1); currentacc += (adc_raw  adc_zero) * (adc_raw  adc_zero); //subtract the zerocurrent value (512 or thereabouts) from the raw readings, square that, and add it to an accumulator variable } currentac = sqrt(currentacc)/samplesnum; //take the square root if the accumulator and divide by the number of readings. for DC current something like this adc_zero = 512; //as read when there is no current on the load sensitivity = 0.04; //as read from datasheet currentdc = (adc_raw  adc_zero) * 5/(sensitivity * 1024) right? Almost. You need to convert (adc_raw  adc_zero) to type 'long' before you square it, because the square may exceed the range of an int. You will need to declare currentac with type 'long' too. After taking the sqrt, you need to adjust it for the sensitivity etc. just as you do for DC. That's a confusing order, I'd apply conversions onebyone:
Just remember that if you apply them one by one, you reduce the ability of the compiler to fold the constants, so there are more floating point operations to do at runtime. That may be easier for debugging, but is not so good for getting good performance.



Logged

Formal verification of safetycritical software, software development, and electronic design and prototyping. See http://www.eschertech.com. Please do not ask for unpaid help via PM, use the forum.




0
Offline
Newbie
Karma: 1
Posts: 35
Arduino rocks


« Reply #7 on: November 19, 2012, 11:11:57 am » 
Almost. You need to convert (adc_raw  adc_zero) to type 'long' before you square it, because the square may exceed the range of an int. You will need to declare currentac with type 'long' too. After taking the sqrt, you need to adjust it for the sensitivity etc. just as you do for DC.
i've left all untyped i know typed code is below. i've tried two ways, but it doesn't seams to works. test 1 subtract the zerocurrent value (512 or thereabouts) from the raw readings, square that, and add it to an accumulator variable. When you have accumulated enough samples, take the square root if the accumulator and divide by the number of readings.
double d = 0; long l = 0; for(;;) { uint16_t i = 0; int16_t adc_raw = 0; int16_t adc_zero = 512; for(i=0; i<100; i++) { adc_raw = adc_read(2); l += pow(adc_raw  adc_zero, 2); _delay_us(100); } d = (sqrt(l)/100) * 5 / (0.04 * 1024.0); l = 0;
test 2 average the square of the current, and take the square root of the average
double c = 0, d = 0; for(;;) { uint16_t i = 0; int16_t adc_raw = 0; int16_t adc_zero = 512; for(i=0; i<100; i++) { adc_raw = adc_read(2); adc_raw = adc_raw  adc_zero; c += pow(adc_raw * 5 / (0.04 * 1024.0), 2); _delay_us(100); } d = sqrt(c)/100; c = 0; ... print c... } i'm testing a 6.4 Amp 220V 50hz device both code i obtain almost the same value, which is 0.27Amp, but is not correct



Logged





0
Offline
Shannon Member
Karma: 206
Posts: 12179
Arduino rocks


« Reply #8 on: November 19, 2012, 11:30:09 am » 
l += pow(adc_raw  adc_zero, 2);
You don't want to use pow() here, it uses float, double and is far too slow (it does logarithms, antilogarithms). long val = adc_raw  adc_zero ; l += val * val ;
You need to be sampling the waveform at 1000Hz or more for good accuracy so its important the maths isn't too slow. Lose the delays  if you want to control the timing you should use millis() or micros() so you can synchronize to a multiple of 50 or 60Hz. First look at a set of raw values from the sensor to check they are what you'd expect (sample a bunch into an array, then print out the array, printing out is too slow also). You should see a clear sinusoidal pattern with the correct amplitude. Then test your calculation code with known correct raw values to see if the arithmetic is correct.



Logged

[ I won't respond to messages, use the forum please ]




0
Offline
Newbie
Karma: 1
Posts: 35
Arduino rocks


« Reply #9 on: November 19, 2012, 02:08:55 pm » 
l += pow(adc_raw  adc_zero, 2);
You don't want to use pow() here, it uses float, double and is far too slow (it does logarithms, antilogarithms). long val = adc_raw  adc_zero ; l += val * val ;
You need to be sampling the waveform at 1000Hz or more for good accuracy so its important the maths isn't too slow. Lose the delays  if you want to control the timing you should use millis() or micros() so you can synchronize to a multiple of 50 or 60Hz. First look at a set of raw values from the sensor to check they are what you'd expect (sample a bunch into an array, then print out the array, printing out is too slow also). You should see a clear sinusoidal pattern with the correct amplitude. Then test your calculation code with known correct raw values to see if the arithmetic is correct. attached a plot of raw values, just analogRead of the pin. it has a strange behavior+ the first part (y=512) is a line, device is power off. + the second part, some sort of 1/2 sine wave, only + part is present, device @ 3.4Amp, [strange here, why??] + the third part is a sine wave, device @ 6.4Amp sensor has the same circuit as the datasheet propose, it does not change if Cf and / or Rf are present or not.



Logged





United Kingdom
Offline
Tesla Member
Karma: 224
Posts: 6614
Hofstadter's Law: It always takes longer than you expect, even when you take into account Hofstadter's Law.


« Reply #10 on: November 19, 2012, 04:44:29 pm » 
What is the device whose current draw you are measuring?



Logged

Formal verification of safetycritical software, software development, and electronic design and prototyping. See http://www.eschertech.com. Please do not ask for unpaid help via PM, use the forum.




Left Coast, CA (USA)
Offline
Brattain Member
Karma: 361
Posts: 17294
Measurement changes behavior


« Reply #11 on: November 19, 2012, 09:11:33 pm » 
What is the device whose current draw you are measuring?
I would ask that also. Current is only drawn in sinewave form if the load is a pure resistance, many loads just 'sip' current when the voltage raises above a certain value. Do an analysis of what AC current input would look like if wired to a typical full wave capacitance filtered DC power supply. The supply only draws current if the DC load wired to the power supply 'pulls' down the voltage stored in the filter capacitors below the peak DC value the caps can charge to. Expecting pure sine wave current flow with real world loads will cause much errors in your 'measurements'. You will need a 'true RMS' calculation derived from the raw sampled measurement values. Lefty



Logged





0
Offline
Shannon Member
Karma: 206
Posts: 12179
Arduino rocks


« Reply #12 on: November 19, 2012, 09:21:37 pm » 
Its either using diode or thyristor to switch to half load?
Either way the RMS calculation should get the right answer.


« Last Edit: November 19, 2012, 09:25:38 pm by MarkT »

Logged

[ I won't respond to messages, use the forum please ]




0
Offline
Newbie
Karma: 1
Posts: 35
Arduino rocks


« Reply #13 on: November 20, 2012, 03:48:12 am » 
it's a 1500w hair dryer with three position (stop, 3.4A, 6.4A, as read by my multimeter, which is not true rms). i've also try attaching another heater, from 6.4A the current sucked it sine wave form BUT, and the extimated current measure is right IF i change the sensitivity A LOT. uint16_t i = 0; double sensitivity = 0.0016; int16_t adc_raw = 0; int16_t adc_zero = 512; //as read when there is no current on the load for(i=0; i<100; i++) { adc_raw = adc_read(2); l += (adc_raw  adc_zero)*(adc_raw  adc_zero); _delay_us(100); } d = (sqrt(l)/100) * 5 / (0.0016 * 1024.0)  0.5; l = 0; d = (sqrt(l)/100) * 5 / (0.0016 * 1024.0)  0.5; with sensitivity of 0.0016 and an offset of 0.5 i've almost right result on 0A and above 3.4A (tried only 6.4A / 8A / 12A) but from datasheet sensitivity is 0.04 ( = 25 * 0.0016)



Logged





United Kingdom
Offline
Tesla Member
Karma: 224
Posts: 6614
Hofstadter's Law: It always takes longer than you expect, even when you take into account Hofstadter's Law.


« Reply #14 on: November 20, 2012, 04:08:18 am » 
1. In the plot you provided in reply #9, at the 6.4A setting, what are the ADC readings at the high and low peaks?
2. How do you know that the current is 6.4A?



Logged

Formal verification of safetycritical software, software development, and electronic design and prototyping. See http://www.eschertech.com. Please do not ask for unpaid help via PM, use the forum.




