Measuring 3- Wire Strain Gauge
Topic: Measuring 3- Wire Strain Gauge
(Read 3 times)
Location: United Kingdom
Hofstadter's Law: It always takes longer than you expect, even when you take into account Hofstadter's Law.
Re: Measuring 3- Wire Strain Gauge
Sep 22, 2011, 09:01 am
*If* the strain gauges are of the sort where one element increases its resistance and the other decreases, then to use all 4 I would connect them like this. Let the 3 terminals of each one be called C (common), I (increase) and D (decrease). Connect them in a ring using the I pins and D pins so that each I pin is connected to another I pin and each D pin is connected to another D pin. Then use the four C pins as the bridge inputs and outputs.
What this gives you is a bridge in which one of the resistors is made up of two I elements in series, its two neigbours are each made up of two D elements in series, and the diagonally opposite resistor is also made up of two I elements in series.
Formal verification of safety-critical software, software development, and electronic design and prototyping. See http://www.eschertech.com. Please do not ask for unpaid help via PM, use the forum.