101, 102, 101, 101, 102, 101, 101, 101.Is this normal for a strain gauge?
Thank you for the advice . I'm going to give it a try tonight. But my question is why i'm I getting those strange values?
You could try putting a low pass filter on the INA125 output
Could also try adding some more caps to the Aref pin to make that cleaner.
since then, I make sure the values are in the sweet spots of calculations. multiplying by 10 or 100 to raise the value has made some numbers easier.
When I take readings it seems to jump around. For example:101, 102, 101, 101, 102, 101, 101, 101.
Would seem a change of much more than 1uV would be needed to be recognized as one level or the other. I would say more on the order of 0.0024 to 0.0025V.Datasheet says:• 0.5 LSB Integral Non-linearity• ± 2 LSB Absolute Accuracy